A residual chlorine meter is a device used to measure the concentration of residual chlorine in water or wastewater. Residual chlorine refers to the amount of chlorine remaining in the water after chlorination, which is important to ensure that the water is safe for consumption or use.
Residual chlorine meters use various methods for measurement, including amperometric titration, colorimetry, and electrochemical sensors. These methods offer different advantages and are suitable for different applications, but all provide accurate and reliable measurements of residual chlorine concentration.
In addition to measuring residual chlorine, some advanced models of residual chlorine meters also include features such as automatic temperature compensation, automatic calibration, and data logging capabilities. These features help to improve the accuracy and reliability of the measurements, as well as provide a convenient and user-friendly experience for the operator.
The SIN-TRC400 residual chlorine meters are commonly used in water treatment plants, swimming pools, and other applications where residual chlorine levels need to be monitored for safety and regulatory compliance. A residual chlorine meter can help ensure that the water being used is safe and free from harmful contaminants, providing peace of mind for both operators and users.
SUP-TRC 400 Residual chlorine meter
|Product||Residual chlorine meter|
|Measure range||Residual chlorine: (0~5) mg/L Temperature: (0.1~40.0)℃|
|Display||7 inch touch screen|
|Protective box size||Dimensions: 400mm×300mm×200mm Window size: 155mm×87mm|
|Transmit output||(4~20) mA (optional)|
|Environmental humidity||≤95% no condensate|
- SUP-ERC 400 Residual chlorine sensor
|Measurement content||HCLO, CLO2|
|Measuring system||Microelectronics MEMS technology, special membrane structure|
|Accuracy||When ≤0.1mg/L, the absolute error is ±0.01mg/L; When ≥0.1mg/L, ±5% of the measured value or ±0.02mg/L (whichever is greater)|
|Polarization time||When using for the first time, first pass water for 2 hours in chlorinated water, and then power on for half an hour.|
|Response time||Less than 30s after polarization is completed|
|Minimum conductivity||≥100 us/cm, can not be used for ultrapure water|
|Operating temperature||(0~40)℃ (non-condensing)|
|Temperature compensation||Pt1000 with built-in integrated automatic compensation|
|Recommended flow rate||≥0.03m/s in flow cell|
|pH range||(5~9) pH, below 5 will damage the membrane head|
|Maximum chlorine concentration||≥5ppm|
|Power supply||Standard 24V DC±2V; optional 12V DC±2V|
|Digital communication||MODBUS RS485|
|Cable length||Standard 3 meters, others can be customized|
|Thread size||NPT 3/4|
|Connection method||5-pin waterproof aviation plug|
|Moisture-proof material||PVC and Viton® O-ring seals|
Get A Quote
If you are interested in our products and want to know more details,please leave a message here,we will reply you as soon as we can.